i-Tear Test (Canada)This page contains information on i-Tear Test for veterinary use.
The information provided typically includes the following:
- i-Tear Test Indications
- Warnings and cautions for i-Tear Test
- Direction and dosage information for i-Tear Test
i-Tear TestThis treatment applies to the following species:
STRIP MENISCOMETRY TUBE
Quantitative tear function test
i-Tear® Test Strip Meniscometry Tube is a rapid 5 second test of the tear function of each eye, using a single-use sterile strip. When the i-Tear® Test is applied to the lateral lower lid tear meniscus of an eye, it absorbs tears via the capillary action of absorber present in the center of the strip. The indicator, placed at the end of the strip, is dissolved in the absorbed tear. After 5 seconds, the length of the stained tear column is marked and read to evaluate tear meniscus volume.
For use under the supervision of a licensed veterinarian. i-Tear® Test is a device applied into a tear meniscus for in vivo examination to assess the tear volume.
1. Configuration Structure: i-Tear® Test is thin strip with a central ditch filled with absorber. Scale is printed on both sides of the absorber to measure the tear volume. Each i-Tear® Test is packaged individually in sterile pouch.
2. Size: Length: 85mm, Width: 7mm, Height: 0.3mm
Indicator: Blue dye 1
Tape: Polyurethane, Polyester
Absorber: Rayon, Pulp
Absorbent area (Tube structure: filled with absorber)
1. Appearance: No damage, no dirt, no foreign material is attached.
2. Sterility assurance level (SAL) 10-6
Box contains 50 individually packed sterile strips. Each strip is used for both eyes.
DEGREE OF ACCURACY
Device error within ±1.2 mm.
DIRECTION FOR USE
1. Carefully open the transparent, sterile pouch at either of the ends of the pouch as to not damage the device.
2. Hold in the center of i-Tear® Test strip as indicated in the adjacent diagram. Do not touch the tips of the strip containing the blue dye area.
3. Apply the tip of the i-Tear® Test of R mark portion on the right eye for 5 seconds to the lateral lower lid tear meniscus without touching the cornea and conjunctiva.
4. Measure the length of the stained tear column, marking by a pen.
5. Repeat the same procedure of L mark portion tip on the left eye.
INTERPRETATION OF THE TEST RESULT
Blue-stained column length reflects the amount of absorbed tear in a proportional manner. The blue-stained column length of approximately 5mm or more is considered as normal, while the length shorter than 5mm is considered as small amount of tear meniscus volume.
i-Tear® Test strip is not meant to be applied to the cornea and conjunctiva. Single use only. Do not re-use.
Do not use if the i-Tear® Test strip is damaged or contaminated. Dispose of it in a safe manner to prevent infection after use. If eye drops have been used, the test should be performed at least 60 minutes later.
For all comments or additional information:
call (514) 685-8118
write to email@example.com
Manufactured in Japan by Echo Electricity Co. Ltd. for: I-MED Animal Health, a division of I-MED Pharma Inc., 1601 St-Régis Blvd., Dollard-des-Ormeaux, QC, Canada H9B 3H7
i-Tear® Test is a registered trademark of I-MED Pharma Inc.
I-MED Animal Health
1601 ST. REGIS BLVD., DOLLARD-DES-ORMEAUX, QC, H9B 3H7
|Every effort has been made to ensure the accuracy of the i-Tear Test information published above. However, it remains the responsibility of the readers to familiarize themselves with the product information contained on the Canadian product label or package insert.|
Copyright © 2019 Animalytix LLC. Updated: 2019-05-31