Accusure Insulin Syringes [31G, 1/2 cc and 1 cc]
Audience: Patients with diabetes mellitus, pharmacists and diabetes healthcare professionalsQualitest Pharmaceuticals and FDA notified patients and healthcare professionals of a voluntary nationwide recall of two lots of Accusure Insulin Syringes. The syringes in these lots have been found to have needles which can detach from the syringe. When the needle becomes detached from the syringe during use, it can become stuck in the insulin vial, push back into the syringe, or remain in the skin after an injection. Consumers who have any recalled Accusure Insulin Syringes (31 G - Short Needle-either 1/2 cc or 1 cc, lot number 6JCB1 or lot number 7CPT1) should stop using them and contact Qualitest at 1-800-444-4011 for product replacement instructions. The lot number can be found on the white paper backing of each individual syringe. These recalled products were distributed from January 2007 through June 2008 to wholesalers and retail pharmacies nationwide (including Puerto Rico).
[08/21/2009- Press Release - Qualitest]
More News Resources
- FDA Medwatch Drug Alerts
- Daily MedNews
- News for Health Professionals
- New Drug Approvals
- New Drug Applications
- Drug Shortages
- Clinical Trial Results
- Generic Drug Approvals
- Monthly Update Archive
Subscribe to our Newsletter
Whatever your topic of interest, subscribe to our newsletters to get the best of Drugs.com in your inbox.