Witness FeLV (Canada)This page contains information on Witness FeLV for veterinary use.
The information provided typically includes the following:
- Witness FeLV Indications
- Warnings and cautions for Witness FeLV
- Direction and dosage information for Witness FeLV
Witness FeLVThis treatment applies to the following species:
FELINE LEUKEMIA VIRUS ANTIGEN TEST KIT
I. General Information
Feline Leukemia Virus (FeLV) is a contagious retrovirus which is endemic particularly among high density or close contact cat populations in many areas of the world. Transmission occurs essentially by contact, mainly through saliva or nasal secretions and by biting or licking. One major source of the virus is persistently viremic, mostly clinically healthy, carrier cats. Vertical (congenital) transmission is also described. A variety of outcomes are possible after cats are exposed to FeLV. Infected cats can become persistently viremic, and many of these cats will die from non-neoplastic diseases. Exposed cats may also develop immunity through the production of virus neutralizing antibodies. Some cats develop an atypical infection where viral replication remains incompletely contained with intermittent antigenemia and occasional shedding of virus. Diagnosis of FeLV infection is usually by detection of the viral antigen, p27, which is in high quantities in the blood of viremic cats.
Ii. Test Indication
WITNESS FeLV is indicated for use when history and/or clinical signs may suggest infection by FeLV or as a screening test prior to an FeLV vaccination. Cats should also be tested prior to entry into FeLV negative households or catteries.
Iii. Test Principle
WITNESS FeLV is a simple test, based on Rapid Immuno Migration (RIM™) technology, which detects the presence of the FeLV antigen p27 in cats’ blood. Sensitized particles bound to p27 antigen present within the sample (whole blood, serum, or plasma) migrate along a nitrocellulose strip. The complex is then captured on a sensitized reaction line where its accumulation causes the formation of a clearly visible pink/purple band. A pink/purple band in the control window ensures that the test was performed correctly.
Iv. Sample Information
- The test can be performed on whole blood, serum, or plasma.
- Whole blood samples must be anticoagulated with EDTA or heparin.
- Samples should be collected with a sterile needle and syringe.
- Hemolysis does not significantly interfere with the test, but strongly hemolyzed samples may partly obscure a weak positive band.
V. Sample Storage
It is recommended to test samples immediately after collection. If samples are kept at room temperature, they should be tested within 24 hours. If testing is further delayed, samples should be refrigerated at 2° to 7°C (35° to 45°F) for up to 7 days. For prolonged storage, samples (serum and plasma only) should be kept frozen at -20°C (-4°F).
Vi. Disposal Of Samples And Test Devices
Handle all samples as if capable of transmitting FeLV. Dispose of samples and used devices appropriately.
Vii. Kit Contents
- 10 pouches, each containing 1 test device.
- 10 disposable pipettes.
- 1 buffer dropper bottle (2.2 ml).
- Instructions for use.
Viii. General Precautions
- Do not use components after expiration date.
- Refrigeration not required. Store the test kit at 2° to 25°C (35° to 77°F). Do not freeze.
- Use the test within 10 minutes after opening the sealed pouch.
- Avoid touching or damaging membrane at windows #1, #2, #3.
- The WITNESS device should be placed on a flat, horizontal surface while performing the test.
- Use a separate pipette for each sample.
- Hold pipette and buffer bottle vertically when dispensing sample and buffer.
- Handle all samples as if capable of transmitting FeLV.
- For veterinary use only.
Ix. Test Procedure And Results
Important: Allow sample and buffer drops to fall onto membrane at window #1. Do not touch pipette tip, sample or buffer drops, or buffer bottle tip directly to the membrane.
1. Sample application
- Tear open the pouch provided and place the test device on a flat horizontal surface for the duration of the test.
- Squeeze the provided pipette near the sealed end. Insert the open end of the pipette into an anticoagulated whole blood, serum, or plasma sample. Release the pressure slightly to draw up a small amount of sample into the pipette.
- Holding the pipette vertically, transfer one drop of sample to the sample well, window #1.
2. Buffer dispensing
- Remove the cap from the buffer bottle, hold it vertically and add two drops of buffer to the sample well window #1.
3. Reading test
- Wait ten minutes, observe the presence or absence of pink/purple bands in reading windows #2 and #3.
- Sample results are read in window #2. The control band is read in window #3.
- The test is complete and may be read before 10 minutes if pink/purple bands are visible in both windows #2 and #3.
- The presence of a pink/purple band only in window #3 before 10 minutes does not mean that the test is complete. A pink/purple band in window #2 may develop slower than the control pink/purple band in window #3.
- Valid test: Test is validated if a pink/purple band is present in the reading window #3.
- Negative for FeLV antigen: No band in reading window #2, with one pink/purple band in window #3.
- Positive for FeLV antigen: One pink/purple band in reading window #2, with one pink/purple band in window #3.
- Invalid test: No pink/purple band in control window #3.
Note: A test result should always be interpreted in the context of all available clinical information and history for the cat being tested.
WITNESS® TEST PROCEDURE
Manufactured by: Synbiotics Corporation, Kansas City, MO 64163 USA
U.S. Vet. License No. 312
NAC No.: 1198506.0
16,740 TRANS-CANADA HIGHWAY, KIRKLAND, QC, H9H 4M7
|Technical Services Canada:||800-461-0917|
|Technical Services USA:||800-366-5288|
|Every effort has been made to ensure the accuracy of the Witness FeLV information published above. However, it remains the responsibility of the readers to familiarize themselves with the product information contained on the Canadian product label or package insert.|
Copyright © 2016 North American Compendiums. Updated: 2016-08-21