Alvedia Feline Quick Test A+B (Canada)This page contains information on Alvedia Feline Quick Test A+B for veterinary use.
The information provided typically includes the following:
- Alvedia Feline Quick Test A+B Indications
- Warnings and cautions for Alvedia Feline Quick Test A+B
- Direction and dosage information for Alvedia Feline Quick Test A+B
Alvedia Feline Quick Test A+BThis treatment applies to the following species:
INSTRUCTIONS FOR THE QUICK TEST PROCEDURE
Quick Test Material
Individual test package contains:
• 1 plastic device with membrane dipstick and 2 empty wells (only 1 well is required to run the test)
• 1 dropper vial of buffer “vet 1” for the dog and “vet 2” for the cat
• 1 blood collector strip (equivalent of 10 µl)
• 1 schematic test manual
• 1 well cover
Same material provided with both Quick Test DEA1.1 and A+B.
• 1 blood typing card to give to the pet’s owner
• 1 blood typing form which will be affixed in the pet’s medical record
Quick Test Procedure
Sample material: EDTA, whole blood, CPD-A, ACD-B, ACD-A, heparin or umbilical cord blood.
For reliable results, use of freshly collected blood is indicated.
Open the dropper vial of buffer, add 3 drops of buffer to the well circled in red (equivalent of 100 µl).
Take the blood collector strip and dip it into the blood tube (direction is indicated by an arrow on the strip) so that the absorbent end is impregnated with blood (a few seconds).
Dip the impregnated end of the collector strip into the well with the buffer (circled in red) and mix gently.
During the mixing, the absorbent end will remain red while the buffer solution also becomes red. After use, discard the collector strip into a suitable waste container.
Hold the device with the wells between thumb and forefingers. With the other hand pull out the membrane holder (having the label).
Insert the legs of the membrane holder into the holes beside the well, so that the membrane end reaches the bottom of the well containing the diluted blood.
Allow to stand until the blood suspension is soaked up by the membrane and the control line (noted “C” on the device) becomes clearly visible.
Click back the membrane holder into the plastic piece with the wells.
Close the well with the well cover.
Break the plastic piece with the wells, then the legs.
You can read the results and store away the flat membrane holder.
Quick Test Results
The dog is DEA1.1 +
Presence of red line in front of “DEA1.1” arrow indicates a positive reaction.
The dog is DEA1.1 -
Absence of red line in front of “DEA1.1” arrow indicates a negative reaction.
The cat is A group
Presence of red line in front of “A” arrow indicates a positive reaction for A group and a negative for B group.
The cat is B group
Presence of red line in front of “B” arrow indicates a positive reaction for B group and a negative for A group.
The cat is AB group
Presence of 2 red lines in front of “A” and “B” arrow indicates a positive reaction for both A and B group.
VERY IMPORTANT FOR ALL QUICK TEST:
The control must always show a positive reaction in front of the “C” arrow, ensuring the test has run successfully.
If not the test must be repeated.
Blood Group Compatibility
In dogs, the determination of DEA 1.1 antigen is strongly recommended before any blood transfusion to avoid a potent alloantibody response against this antigen and to avoid acute haemolytic transfusion reaction.
In cats, the presence of naturally occurring alloantibodies in type A and in type B cats requires that blood typing be performed prior to blood transfusion to avoid acute haemolytic transfusion reaction, and in breedings to prevent neonatal isoerythrolysis.
Alvedia, Lyon Bioparc, 60 avenue Rockfeller, 69008 Lyon, France
Phone: + 33 478 380 239
Fax: + 33 478 939 563
NAC No.: 1626000.1
AB71 - 2055 NOTRE DAME AVENUE, WINNIPEG, MB, R3H 0J9
|Every effort has been made to ensure the accuracy of the Alvedia Feline Quick Test A+B information published above. However, it remains the responsibility of the readers to familiarize themselves with the product information contained on the Canadian product label or package insert.|
Copyright © 2016 North American Compendiums. Updated: 2016-10-31