Assure/FeLV (Canada)This page contains information on Assure/FeLV for veterinary use.
The information provided typically includes the following:
- Assure/FeLV Indications
- Warnings and cautions for Assure/FeLV
- Direction and dosage information for Assure/FeLV
Assure/FeLVThis treatment applies to the following species:
Feline Leukemia Virus Antigen Test Kit
For the Detection of Feline Leukemia Virus
Feline Leukemia Virus (FeLV) is a highly contagious oncogenic RNA virus that causes both neoplastic and non-neoplastic diseases in cats. Diseases caused by FeLV include lymphosarcoma, myelogenous leukemia, thymic atrophy, nonregenerative anemia and panleukopenia-like disease. Because FeLV is immunosuppressive, it predisposes infected cats to a variety of secondary diseases.
Rapid identification of infected animals allows them to be separated from non-infected cats to prevent the spread of FeLV. ASSURE®/FeLV uses highly specific antibodies to quickly identify FeLV-infected cats. The group specific antigen, p27, is found in high levels in infected cats and
its presence is diagnostically definitive for FeLV infections. ASSURE®/FeLV is an immunoenzymatic assay using antibodies that specifically recognize p27 in cat blood and saliva.
Following exposure, a cat may test positive within 14 days. Persistent infections will remain positive while transient infections may turn negative within 1 to 2 months. A saliva test will not be strongly positive until the virus is amplified in the bone marrow, released into the circulation, and finally has infected the epithelial tissues. This could take a period from 1 to 2 months following exposure and would indicate a persistent infection.
Since infection may be transient in cats that develop immunity, antigen-positive animals should be retested in 1 to 2 months. A positive second test indicates persistent infection. A negative test indicates clearance of the virus.
Ii. Test Principles
The bulbous ends of the plastic wands are coated with an antibody directed against a specific FeLV antigen, p27. A second antibody directed against the p27 antigen is conjugated to the enzyme horseradish peroxidase. The sample is incubated in a single step with both the antibody-coated wand and enzyme-labeled antibodies. If antigen is present, it is captured by the wand. The enzyme-labeled antibodies are in turn captured by the antigen on the wand. The unbound enzyme labeled antibody is washed away and the wand placed into a chromogenic substrate.
The development of a distinct blue color in the solution indicates the presence of FeLV antigen.
In the absence of FeLV, no color will develop.
When a saliva sample is used, ASSURE®/FeLV can detect antigen in cats actively shedding virus within 15-20 minutes.
When whole blood, serum or plasma samples are used, ASSURE®/FeLV can detect circulating antigen in infected cats within 10-15 minutes.
Iii. Common Leukemia Diagnostic Questions And Answers
How soon after exposure/infection can antigen be detected?
As early as 4 days, but 14 days is average. It transmitted in-utero, antigen could be present and detected at birth.
Will FeLV vaccination cause the test to be positive?
No. ASSURE®/FeLV tests for FeLV p27 antigen in blood or saliva. Vaccination stimulates antibody production with a minute amount of p27; insufficient for detection. FeLV p27 antigen can only be found in detectable amounts when a cat is infected with the virus.
Is blood better to use for testing than saliva?
A positive result with a blood sample could indicate a persistent or transient infection. A positive result with a saliva sample is indicative of a persistent viremia. Positive blood samples
can be repeated using saliva to determine the state/stage of the disease.
What does an ELISA positive, IFA negative test imply?
The infection is not yet established (i.e. amplified in the bone marrow) and may become transient, latent or persistent. Retesting in 1 to 2 months is recommended.
IV. CONTENTS OF ASSURE®/FeLV TEST KIT
Anti-FeLV Antibody Coated Wands
Predispensed HRP-Monoclonal Antibody Conjugate (A Tubes)
Predispensed Substrate Buffer (B Tubes)
Bottle C-Chromogen (White Cap)
Bottle D-S.E.R. (Blue Cap)
Materials required, but not provided:
Distilled/deionized water or normal saline
1. Allow kit to come to room temperature (21°-25° C; 70°-78° F) prior to use; approximately one hour.
2. Do not expose kit to direct sunlight.
3. Do not use expired reagents or mix from different kit serials.
4. Hold reagent vials vertically for proper drop volume.
5. Dispose of potentially infected specimens appropriately.
6. FOR VETERINARY USE ONLY.
Vi. Sample Information
Saliva or 100 µI (0.10 ml) of whole blood, plasma or serum is required. Whole blood and plasma must contain an anticoagulant. Serum and plasma samples may be stored at 2°-7°C (36°-45°F) for up to 7 days; 24 hours for whole blood. For longer periods, serum and plasma (not whole blood) may be stored at -20°C. Samples may be run cold. Hemolyzed or lipemic samples can be used, however, severely hemolyzed or lipemic samples may produce background color. When in doubt, obtain a better quality sample. Saliva samples cannot be stored or preserved. Saliva cell debris or abrasions on the wand’s surface will not affect test results. Do not chemically induce salivation.
Vii. Storage And Stability
Store the test kit at 2°-7°C (38°-45°F). Do not freeze. Reagents are stable until expiration date provided they have been stored properly.
1. Hardy, W.D., Jr., 1974 Veterinary Clinics of North America 4:133-146.
2. Hardy, W.D., Jr. McClelland, A.J., Zuckerman, E.E., Hess, P.W., Essex, M., Cotter, S.M., MacEwan, EIG., and Hayes, A.A., 1976, Comps. Leuk. Research (Karger, Basel) 43:511-514
3. Lewis, M.G., Wright, K.A., Lafrado, L.J., Shanker, PIJ., Palumbo, N.E., Lemoine, E.D., and Olsen, R.G., 1987, J. Clin. Micro. 25:1320-1322.
4. Ettinger, S.J.: Textbook of Veterinary Internal Medicine. Philadelphia: W.B. Saunders Co., 1959, pp. 328-329.
Assure®/felv Test Procedure
NOTE: Use WHOLE BLOOD, SERUM, PLASMA or SALIVA samples.
Prior to use, allow components to come to room temperature (70° to 78°F; 21° to 25°C).
For each sample you will need:
Anti-FeLV Antibody Coated Wand. Label with cat’s ID.
Transfer Pipette (one use only) 3 drops = 100 µl (Not required for saliva samples.)
Predispensed HRP-Monoclonal Antibody Conjugate Tube A.
Predispensed Substrate Buffer Tube B.
Workstation with Reagents.
Squirt Bottle with Distilled or Deionized Water or Saline.
A. SAMPLE COLLECTION
1A. WHOLE BLOOD/SERUM/PLASMA PROCEDURE:
Pipette 3 drops sample to A Tube.
Tap to mix.
Proceed to Step 2.
1B. SALIVA PROCEDURE:
Add 3 drops Bottle D (Blue Cap) to A Tube.
Tap to mix.
Set in workstation.
- Obtain labeled Wand.
- Place bulbous end of Wand between cheek and gum in the back of the mouth.
- Gently rotate for 5-10 seconds.
Proceed to Step 2.
B. CONJUGATE INCUBATION
2. Place bulbous end of labeled Wand in A Tube.
Twirl 1-3 seconds to mix.
WAIT 5 MINUTES for blood samples,* or
WAIT 10 MINUTES for saliva samples.*
*Or up to 20 minutes.
C. PREPARE B TUBE
3. During waiting period:
Remove stopper from B Tube.
Add 3 drops Bottle C (White Cap) to B Tube.
Tap to mix.
Set aside for use in Step 5.
D. WASH STEP
4. NORMAL SALINE MUST BE USED WITH WHOLE BLOOD SAMPLES.
Deionized/distilled water or normal saline can be used with serum, plasma or saliva samples.
Remove Wand from A Tube.
Wash bulbous end and tip by using METHOD I or II:
I. SQUIRT BOTTLE METHOD
- Direct a forceful stream of the appropriate washing agent against bulbous end and tip of Wand and work up handle.
- Shake off excess water.
- Repeat 5-7 times totaling approximately 8 oz. of appropriate washing agent.
II. CUP METHOD
- Perform initial rinse of Wand with a forceful stream of appropriate washing agent.
- Swirl/Swish vigorously in 100 ml of appropriate washing agent for a minimum of 30 seconds.
- Shake off excess liquid.
- Replenish liquid between Wands.
E. COLOR DEVELOPMENT
5. Place washed Wand in B Tube.
Twirl 1-3 seconds to mix.
WAIT 5 MINUTES. (Weak positives may be verified by waiting up to 10 minutes).
6. Observe solution against workstation window or a white background for blue color.
NOTE: Color intensity may vary with level of FeLV antigen present.
Optional Procedural Control
To verify technique and kit performance when a negative/clear result is obtained:
Place wand back into A Tube.
Twirl to mix for 1-3 seconds.
Do not wash.
Place back into B Tube.
Blue color will develop within 1 minute indicating reagents were added correctly and kit is performing properly. If color does not develop, repeat the test. (This is a procedure and reagent check only. FeLV antigen is not present.)
Good Techniques = Accurate Results
Whole blood and plasma must contain anticoagulant.
Hemolyzed and lipemic samples may be used, however, normal saline should be used in place of distilled or deionized water in step 4. Severely hemolyzed and lipemic samples may produce background color. When in doubt, obtain a better quality sample.
Washing is the most important step. Wands cannot be overwashed. Underwashing will result in non-specific blue color development in the B Tube.
Prolonged incubation for more than 10 minutes in step 5 may result in non-specific blue color development. Read results at 5 minutes. If no color is seen at 5 minutes, the sample is negative. Weak or suspect positives at 5 minutes may be verified by waiting up to 10 minutes.
Do not use the test kit past the expiration date and do not intermix components from different serial numbers.
Store kit at 2° to 7°C (36° to 45°F). Allow kit to come to room temperature before use.
FOR TECHNICAL ASSISTANCE: 1-800-228-4305
SYNBIOTICS CORPORATION, 16420 Via Esprillo, San Diego, CA 92127
U.S. Veterinary License Number 312
ASSURE®/FeLV is a registered trademark of Synbiotics Corporation.
©1994 Synbiotics Corporation
NAC No.: 12550091
Distributed by ZOETIS CANADA
16,740 TRANS-CANADA HIGHWAY, KIRKLAND, QC, H9H 4M7
|Technical Services Canada:||800-461-0917|
|Technical Services USA:||800-366-5288|
|Every effort has been made to ensure the accuracy of the Assure/FeLV information published above. However, it remains the responsibility of the readers to familiarize themselves with the product information contained on the Canadian product label or package insert.|
Copyright © 2015 North American Compendiums. Updated: 2015-04-03