Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot TestThis page contains information on Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot Test for veterinary use.
The information provided typically includes the following:
- Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot Test Indications
- Warnings and cautions for Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot Test
- Direction and dosage information for Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot Test
Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot TestThis treatment applies to the following species:
One-step Canine Parvovirus Antigen Test
The Canine Parvovirus Antigen Test Kit, Blot Test is a chromatographic immunoassay for the qualitative detection of Canine Parvovirus antigen in canine feces.
The Canine Parvovirus Antigen Test Kit, Blot Test has a letter “T” and “C” as test line and control line on the surface of the device. Both the test line and control line in the result window are not visible before applying any sample. The control line is used for procedural control. The control line should always appear if the test procedure is performed properly and the test reagents of the control line are working. A purple test line will be visible in the result window if there is Parvovirus antigen in the specimen. The specially selected Canine Parvovirus antibodies are used in the test band as both capture and detector materials. These enable the Canine Parvovirus Antigen Test Kit, Blot Test to identify Canine Parvovirus antigens in canine feces with a high degree of accuracy.
- Materials provided (5 or 10 tests/kit)
1) Five (5) or ten (10) Anigen Rapid Canine Parvovirus Antigen Test Devices
2) Five (5) or ten (10) Specimen tubes containing assay diluent buffer
3) Five (5) or ten (10) Sample collection swabs
4) Five (5) or ten (10) Disposable droppers
5) One (1) Instruction for use
1) For veterinary diagnostic use only.
2) For best results, strict adherence to these instructions is required.
3) All specimens should be handled as being potentially infectious.
4) Do not open or remove test devices from their individually sealed pouches until immediately before their use.
5) Do not use the test device if the pouch is damaged or the seal is broken.
6) Do not reuse test device.
7) All reagents must be at room temperature before running the assay.
8) Do not use reagents beyond the stated expiration date marked on the label.
9) The components in this kit have been quality control tested as standard batch unit. Do not mix components from different lot numbers.
10) Do not mix components from different lot numbers.
11)The assay diluent contains sodium azide, a hazardous chemical. Dispose of used product by wrapping with paper and putting in garbage.
- Storage And Stability
The kit can be stored at room temperature or refrigerated (2-30°C) or refrigerated. The test kit is stable through the expiration date marked on the package label. DO NOT FREEZE. Do not store the test kit in direct sunlight.
- Specimen Collection and Preparation
1) The samples from canine feces should be used for this test.
2) The specimens should be tested immediately as soon as the samples are collected.
- Procedure of the test
1) Collect the samples from canine feces using the swab.
2) Insert the swab into the specimen tube containing 1ml of assay diluent.
3) Mix the swab samples with the assay diluent to extract well.
4) Remove the test device from the foil pouch, and places it on a flat and dry surface.
5) Use the disposable dropper provided to take the samples from extracted and mixed specimens in the tube.
6) Add four (4) drops into the sample hole using the disposable dropper.
The mixed assay diluent should be added exactly, slowly, drop by drop.
7) As the test begins to work, you will see purple color move across the result window in the center of the test device. If the migration has not appeared after 1 minute, add one more drop of the mixed assay diluent to the sample well.
8) Interpret test results at 5 - 10 minutes. Do not decide after 20 minutes.
- Interpretation Of The Test
A color band will appear in the left section of the result window to show that the test is working properly. This band is the control band “C”. The right section of the result window indicates the test results. If another color band appears in the right section of the result window. This band is the test band “T”.
1) Negative Result
The presence of only one band “C” within the result window indicates a negative result.
2) Positive Result
The presence of two color bands (“T” and “C”) within the result window, no matter which band appears first indicates a positive result.
3) Invalid Result
If the purple color band is not visible within the result window after performing the test, the result is considered invalid. The directions may not have been followed correctly or the test may have deteriorated. It is recommended that the specimen be retested.
- Limitations Of The Test
Although the Canine Parvovirus Antigen Test Kit, Blot Test is very accurate in detecting Canine Parvovirus antigen, a low incidence of false results can occur. Other clinically available tests are required if questionable results are obtained. As with all diagnostic tests, a definitive clinical diagnosis should not be based on the results of a single test, but should only be made by the veterinarian after all clinical and laboratory findings have been evaluated.
- Key for symbols on box:
In vitro diagnostic
Store between 2-30°C
Read insert for full information.
Doc. No.: I 1101-02US
Issued date: Oct 25, 2012
Manufactured by BioNote, Inc., 2-9, Seogu-dong, Hwaseong-si, Gyeonggi-do, Korea (445-170)
Tel: +82-31-211-0516, Fax: +82-31-8003-0618,
Manufactured for Modern Veterinary Therapeutics, LLC, 18001 Old Cutler Road, Suite 317, Miami, Florida 33157 - USA
Tel.: +1 305 234 8228 Fax.: +1 305 503 8585
U.S. Permit No. 544B
NAC No.: 13050051
18001 OLD CUTLER ROAD, SUITE 633, MIAMI, FL, 33157
|Every effort has been made to ensure the accuracy of the Anigen Rapid Canine Parvovirus Antigen Test Kit, Blot Test information published above. However, it remains the responsibility of the readers to familiarize themselves with the product information contained on the US product label or package insert.|
Copyright © 2014 North American Compendiums. Updated: 2013-12-23